Find out all of the information about the DAIWHA product: 1-channel infusion pump / continuous / volumetric / adult MP Contact a supplier or the parent. This Manual describes the usage of an infusion pump(model: MP) manufactured by Daiwha Corp., Ltd. Only the person who is. Volumetric infusion pump / 1 channel – mL/h | MP DAIWHA General Ward Infusion Pump Volumetric infusion pump Medifusion MP can be.
|Published (Last):||15 October 2007|
|PDF File Size:||13.12 Mb|
|ePub File Size:||10.62 Mb|
|Price:||Free* [*Free Regsitration Required]|
Bomba Infusion Mp – [PDF Document]
If any infusion set other than specified is used, proper flow rate accuracy and alarm function air and occlusion are not guaranteed. Option of alarm sound is low 1. Operate the pump by the designated steps.
The motor rotation detector assures that pump drive rotation occurs or the motor shuts down.
Battery operating time 1. Display data controller is 1. Just order what you want.
Here is the right place! Ascom appoints Ljubisav Matejevic to lead the Strategic Alliances business Ascom invests further into its strategic partners business by appointing Check both pump and patients conditions on a regular basis while drugs are infused. Only Stop key is usable 3. Leakage current should daiqha exceed 1mA. Same causes as 3.
1-channel infusion pump / continuous / volumetric / adult – MP – DAIWHA
Malfunction in door circuit 1. Check power switch off, the pump is not off. If you do not follow the instruction, it can cause minor injuries or damage on the product. Use recommended infusion detected.
High Quality For DAIWHA MP 1000 Battery | Replacement For DAIWHA MP-1000 Syringe Pump Battery
It may therefore be necessary to recharge the battery after a storage period. Backlashing of motor 1.
Convenient battery indicator 7. Display data controller is 2. Post on Oct views.
MP-1000 > DISCONTINUED
You want it, we got it. The batteries supplied by our Company if not mark as [Original] are [replacement for] sold for use with certain products of computer manufacturers, and any reference to products or trademarks of such companies is purely for the purpose of identifying the computer manufacturers with which our products [are replacement for] may be used. Check R12, R14, R17 3. Volume display window 1.
Have a good shopping, Have a good day! Check Motor and replace 6. Reload infusion set 3.
Check display segment is abnormal. When usable at 5. Abnormal pumping unit 3.
To discharge, simply run your device under the battery’s power until it shuts down or daiwhha you get a low battery warning. Abnormal battery capacity 4. Malfunction of buzzer circuit 2. Properly attach the pump.